This document applies only to static watt-hour meters of accuracy classes A, B and C for the measurement of alternating current electrical active energy in 50 Hz or 60 Hz networks and it applies to their type tests.
NOTE 1 For general requirements, such as construction, EMC, safety, dependability etc., see the relevant EN 62052 series or EN 62059 series.
This document applies to electricity metering equipment designed to:
- measure and control electrical energy on electrical networks (mains) with voltage up to 1 000 V AC;
NOTE 2 For AC electricity meters, the voltage mentioned above is the line-to-neutral voltage derived from nominal voltages. See EN 62052-31:2016, Table 7. EN 62052 31:2016 covers AC voltages only up to 600 V and Ed. 2 of EN IEC 62052 31 will cover AC voltages up to 1000 V.
- have all functional elements, including add-on modules, enclosed in, or forming a single meter case with exception of indicating displays;
- operate with integrated or detached indicating displays;
- be installed in specified matching sockets or racks;
- optionally, provide additional functions other than those for measurement of electrical energy.
Meters designed for operation with low power instrument transformers (LPITs as defined in the EN 61869 series) can be tested for compliance with this document only if such meters and their LPITs are tested together and meet the requirements for directly connected meters.
NOTE 3 Modern electricity meters typically contain additional functions such as measurement of voltage magnitude, current magnitude, power, frequency, power factor, etc.; measurement of power quality parameters; load control functions; delivery, time, test, accounting, recording functions; data communication interfaces and associated data security functions. The relevant standards for these functions could apply in addition to the requirements of this document. However, the requirements for such functions are outside the scope of this document.
NOTE 4 Product requirements for power metering and monitoring devices (PMDs) and measurement functions such as voltage magnitude, current magnitude, power, frequency, etc., are covered in EN 61557-12:2008. However, devices compliant with EN 61557-12:2008 are not intended to be used as billing meters unless they are also compliant with the EN IEC 62052-11:2021/A11:2022 and EN 50470-3:2022 standards.
NOTE 5 Product requirements for power quality instruments (PQIs) are covered in EN 62586-1:2017. Requirements for power quality measurement techniques (functions) are covered in EN 61000-4-30:2015. Requirements for testing of the power quality measurement functions are covered in EN 62586-2:2017.
This document does not apply to:
- meters for which the line-to-neutral voltage derived from nominal voltages exceeds 1 000 V AC;
- meters intended for connection with low power instrument transformers (LPITs as defined in the EN 61869 series) when tested without such transformers;
- metering systems comprising multiple devices (except of LPITs) physically remote from one another;
- portable meters;
NOTE 6 Portable meters are meters that are not permanently connected.
- meters used in rolling stock, vehicles, ships and airplanes;
- laboratory and meter test equipment;
- reference standard meters;
- data interfaces to the register of the meter;
- matching sockets or racks used for installation of electricity metering equipment;
- any additional functions provided in electrical energy meters.
This document does not cover measures for the detection and prevention of fraudulent attempts to compromise meter’s performance (tampering).
When the standard is harmonized under a directive or regulation, the word Harmonized appear. When such an inscription does not appear, it means that the standard is related to / supporting the relevant directive or regulation.
Directive 2014/32/EU of the European Parliament and of the Council of 26 February 2014 on the harmonisation of the laws of the Member States relating to the making available on the market of measuring instruments